sexual disfunctional therapy Equipment
Repeat the abovementioned therapy process some times. After ending
therapy by the sexual disfunctional therapy Equipment, the
instrument sounds alarm and is automatically stopped. At this
moment, time display is 00. Turn negative pressure adjustment knob
counterclockwise to end.
During the course of therapy, press start key for suspending
therapy; again press start key for next therapy; Press set key for
Notice for the sexual disfunctional therapy Equipment:
During the course of therapy, Sustained negative pressure value is
not greater than 0.025 Mpa. Too large negative pressure may cause
hydrops or subcutaneous bleeding. Ask the patient's own feeling and
adjust negative pressure properly. Record all results of the
treatment, such as the feeling of patients etc.
Operating essentials for thesexual disfunctional therapy Equipment:
A) Therapy strength is determined according to patient’s bearing
B) Before treatment patients should shave pubic hair entirely to
ensure negative pressure attraction effect
C) A small number of patients with prolonged foreskin may occur
foreskin edema; we should decrease negative pressure strength
properly during therapy.