Installation of the ED equipment China
Notice: The instrument shall be used in clean room with adequate
air circulation. Its environment is clean and dry. Don’t install
the instrument in a location near chemical materials and electric
devices with strong electromagnetic and electric fields. The
instrument position should easy in operation.
After unpacking, check all the parts according to packing list.
Host computer and accessories shall be in conformity with packing
list. Read this operation manual carefully and know using method of
the instrument well.
Connect the monitor.
Insert earphone and foot switch into the place which fixed on the
Mount the bracket of negative pressure pipe
Insert D type plug of electrode conductor (with multiple output
ends) into electric frequency socket on the rear panel and turn
Connect the joint at the bottom of negative pressure pipe with
one end of attraction hose; connect other end of the hose with
filter network. Connect filter network to negative pressure outlet
trough rubber tube and turn tightly.
Connect one end of the other attraction hose with the joint at
the side of negative pressure pipe and the other end of that with
bubble massage outlet.
Check power supply source for the instrument and determine it
conforms to given power supply parameters. After this, insert one
end of power wire into the place with 220V/50Hz on the rear panel
of host computer and connect the other end of that into connection
board for power supply.
Check all connecting wires of the instrument and then electric
power is turned on.
Notice of the ED equipment China:
1. Do not pull or insert electrode plug under on-position to avoid
damaging electrodes and host computer.
2. Must not use diagnosis function and therapy function
simultaneously, avoiding being damaged.
3. Must install filter network. Otherwise, impurity will clog
solenoid valve and disrupt normal operation. Should clean and
sterilize filter network regularly