Model: MCV-400A (Without air compressor)
This Neonatal Ventilator is a ventilation device for infant with
novel structure and excellent performance,which adopts opening
ventilation mode with high frequency ,small tidal volume and low
Feature of the Neonatal Ventilator :
A. The tidal gas can be exported with constant pressure, contact
current and accurately adjustable flow.
B. Opening ventilation, which used opening intubation without
casing bag, and get little resistance when expiratory is suitable
for the ventilation management of infant .
C. Evident effectiveness of ARDS treatment can be shown with
inserting small tube and jetting oxygen through nasal vestibule,
which is safe and doesn't cause pressure injury easily.
D.Special CPAP tube for infant ventilation treatment are available
E.PEEP and humidifier volume can be adjusted accurately.
F. Alarm system with sound and light when no oxygen, power failure
or low airway positive pressure.
G. Heated humidifier can be used for good effectiveness of
H. Inbuilt storage battery, it can be used while being charged and
can be used for more than four hours after full charged .
F. It is suit for these neoate ,infant ,pediatric whom weight is
Main Technique Parameter of the Infant Ventilator:
I/E ratio:1:1～ 1:3 and inverse ratio
Driving Pressure :0～60KPa
Humidifying rate :≥6ml/h
Process Ventilation :T=10min,
ProI : T1(150bpm):T2(35bpm)=7:3
ProII: T2(35bpm): T1(150bpm)=7:3
Storage Battery: 4Ah, 12V, Rechargeable
Alarm: Alarm with sound and light when lack of power, or the airway
pressure is over 1Kpa±0.2Kpa and alarm will last for 6 seconds when
Oxygen Concentration :21%-90%(with air compressor) ; 40%-90%(with
Ventilation Mode: HFJV(High Frequency Jet
Ventilation);CPAP(Continuous positive airway
pressure);HFJV+PEEP(Positive End-Expiratory Pressure),
HFJV+SIGH(Sigh), Pro I and Pro II ventilation . Manual ventilation.